a) Connect pump to 240 volt power supply.
b) Place medication into nebulizer bowl.
c) Attach one end of the tubing to the nebulizer bowl and the other
end to the pump.
d) Fit the mask or the mouth piece to the top of the nebulizer bowl
and place the mast on your face or the mouth piece in your mouth.
e) Switch on the pump and breathe normally through your mouth.
When the bowl has stopped nebulizing turn the unit off.
DYNAMIC FLOW WITH NEBULIZER
SOUND LEVEL AT 1 METRE
AVERAGE RATE (nebulizer)
240 V 50 HZ
275 X 200 X 100 MM
.5 - 5 MICRON
1 ML / 5 MIN
L.R. INSTRUMENTS PTY LTD
RINGWOOD VIC 3134
Warranty claims or service requirement enquiries should be directed to the
place of purchase or to:
L.R. INSTRUMENTS PTY LTD
(03) 9870 7066
1800 337 048
Does not cover accessories, filters or damage due to misuse or negligence.
Home Use: 5 year warranty
Hospitals and Medical Centres: 2 year warranty
CONTACT PHONE NUMBER A.H. _____________________
a) After each use, dismantle the nebulizer bowl for cleaning in warm
b) Wash the mask, mouth piece and bowl in warm water. (Never boil your
accessories). Rinse and place to side to dry.
c) Do not wipe your nebulizer bowl with a cloth or tissue as lint may lodge
in the jets.
d) Replace your nebulizer bowl every 4-6 months depending on use and
e) Wipe spills off pump housing with a damp cloth.
SERIAL NUMBER: _________________________
a) Check the age of your bowl if it is frequently used (i.e.. 4 times per
day). It should be replaced every 4-6 months.
b) Check inlet filter for blockage or dicolouration. If it is discoloured,
replace with a new one from your BESTNEB stockist.
c) Make sure air supply tubing is not kinked or broken.
d) Check that the centre jet in the nebulizer is not blocked or
damaged. If it is, soak it in water and clean thoroughly. Do not
push a pin into the jet as this will cause damage. If jet is damaged,
replace complete bowl.
e) Ensure that centre jet is in place in your bowl.
If the bowl is taking a long time to nebulize your medication or is
NEBULIZER BOWL PROBLEM SOLVING
Do not use the pump on a bed, on carpet or on a soft surface, as the
units air vents may become blocked.
Do not block the air vents at any time during operation.
Do not operate the pump if the power cord is frayed or damaged in
Do not use where oxygen or other flammable gases are present.
Keep your pump clear of water at all times.
Electrical products that fall into water should be turned off and the plug
removed from the power point before they are retrieved. Do not store
or use this unit near water.
INSTRUCTION AND WARRANTY CARD
PLEASE COMPLETE IF RETURNING UNIT UNDER A WARRANTY CLAIM
PROOF OF PURCHASE MUST ACCOMPANY ANY WARRANTY CLAIM
__________________________ STATE: _________ POST CODE: ____________
DATE OF PURCHASE: __________________
PLACE OF PURCHASE: ____________________________________________________________