AUTO LOGIC / AURA LOGIC
Quick Reference Guide
This Quick Reference Guide has been produced to assist Healthcare Professionals and other users with only the functional aspects
of the system. The guide does not replace the general safety warnings, cautions and recommendations provided in the product
instructions for use, part number 630933EN.
Refer to the product instructions for use before using this product.
The Auto Logic® and Aura Logic® mattresses are designed for patients weighing up to 200 kg (440 lb).
The Auto Logic and Aura Logic systems are indicated for the prevention and/or management of all categories of pressure ulcer ,
when combined with an individualised, comprehensive pressure ulcer protocol.
Contraindication: Do not use Auto Logic and Aura Logic systems for patients with unstable spinal fractures.
Caution: If patients have other unstable fractures, or conditions which may be complicated by a soft or moving surface, advice should
be sought from an appropriate clinician before use.
The red Low Pressure indicator is illuminated whenever the pump detects low pressure within the mattress/seat.
An audible alarm sounds unless cancelled by the mute button. The indicator extinguishes once normal pressure is
reached. Check CPR is closed and tubeset is connected properly.
The yellow Battery Low indicator illuminates two hours before battery failure. One hour before battery failure, the pump
The red Power Fail indicator illuminates when a mains power failure has been detected and no battery backup is
available. An audible alarm sounds until power is resumed or the pump is switched off using the Run/Standby button.
The yellow Service indicator illuminates and remains on after a set number of running hours; this indicates that the
pump is ready for a service. The pump continues to function normally even when the service indicator is illuminated. If
During an alarm condition (except Power Fail Alarm), the sound of the alarm can be muted by pressing this button.
The yellow indicator then remains on but the alarms are muted for 15 minutes or until the alarm condition has been
Selects the Reactive (Constant Lower Pressure) (CLP), non-alternating mode. The yellow indicator illuminates when
the pump is in this mode. When the Active (Alternating) mode (default) is selected, the yellow indicator is switched off.
Press this button for approximately 4 seconds to activate the Reactive (CLP) Mode. The change in the mode will
occur at the end of the next complete alternating cycle, and so a maximum 10 minute delay may be experienced.
Control LEDs are illuminated. This operation will not function whilst a seat cushion is connected.
The mattress/seat cell pressure can be manually adjusted for patient comfort using the buttons on the pump control
panel. The pressure setting is indicated by the green indicator to the left of the buttons.
(Part No. BBP600)
CAUTION: The battery pack is not
suitable for use in the homecare
1. NPUAP/EPUAP International Pressure Ulcer Guideline, 2009.
(Part No. PXS001)
Auto Logic 110 (Mattress Overlay)
Place the overlay on top of the base
mattress, with the tube-set located near
the foot end of the bed and the CPR at
the head end. The cells of the mattress
must be uppermost.
to the tubeset and CPR.
Run the straight part of the cable along the side of the mattress
Secure the overlay to the base mattress
using the four corner straps.
Auto Logic 175/200 (Mattress Replacement)
Place the Mattress Replacement directly
onto the bed base, with the tube-set located
near the foot end of the bed and the CPR
at the head end. The cells of the mattress
must be uppermost.
Use the straps provided (4 on each side) to ensure that the MR
is secured to the moving parts of the bed platform.
To Complete the Mattress Installation
Zip the protective cover over the mattress.
Ensure that the logo is uppermost and at
the foot end of the mattress.
Ensure that the CPR unit is clicked into the
Ensure CPR unit and power cord are clear
other possible entrapment areas. The CPR
must be accessible at all times.
Aura Logic Seat Cushion
Place the cushion on top of the chair
surface. From a standing position in
front of the chair and facing it, ensure
that the cells are uppermost, the tubeset appears from the front right corner
of the cushion and the cells are in a
horizontal position across the chair
with the ‘V’ shape pointing towards
and always use the seat cushion with the protective top cover.
Regarding the use of safety sides, always refer to your local
trust policy AND the product instructions for use.
When the Auto Logic pump is switched on at the mains supply
or a battery pack is connected, an audible beep will sound and
a self-diagnostic check will run for approximately 10 seconds.
When the check is complete a second audible beep will sound
and the pump is ready for use.
Press the Run/Standby button on the control
panel of the pump.
Allow approximately 7 minutes for the mattress overlay, 15
minutes for the mattress replacement or 3 minutes for the seat
Place a bed sheet over the mattress and tuck in loosely. Once
the patient is on the mattress, the pump will automatically sense
and adjust the pressure in the cells to support the patient.
To transport a patient using the Auto Logic mattress system
press the Run/Standby button for 2 seconds to set the pump to
standby and then disconnect the tube-set from the pump. This
automatically switches the system into transport mode. The
patient remains supported by the mattress for up to 12 hours.
IN THE EVENT OF CARDIAC ARREST
In the event of a patient suffering cardiac arrest and CPR
needing to be administered:
Installing the Pump
Position the pump, feet down, on
any convenient horizontal surface or
alternatively suspend from the bed foot
rail by means of the swing out hooks.
To Activate the CPR:
Press the two CPR release
Ensure that the mattress/seat tube-set
is not “kinked” or twisted and connect it
to the pump until it clicks into place. Ensure that the tube-set is
securely connected to the pump.
Insert the mains power plug into a suitable mains power socket.
Hans Michelsensgatan 10
211 20 Malmö, Sweden
To Reset the CPR:
Push the front of the CPR
unit until it clicks into place.
® and ™ are trademarks of ArjoHuntleigh
As our policy is one of continuous improvement, we reserve the right to modify designs without prior notice.