QUICK START GUIDE FOR SYNERGYRF AR-9800 UNPACKING: Carefully unpack and inspect all components for shipping damage. Save the packaging for later transport of the device. The footswitch is provided separately. The probe is provided separately and is in sterile packaging. Users must read the SynergyRF User’s Manual and ApolloRF™ Probe DFU prior to using these devices. The User Manual and DFU provide the indications and contraindications for use, warnings, symbol definitions and the full list of product instructions. These documents can be found on Arthrex.com. Setup
Footswitch Cable Connector
Power Cord Generator Probe Connector Probe Suction Tube Footswitch
1 Place console on a flat, dry surface 7 For suction probes, attach suction tube
2 Connect power cord to console 8 Attach footswitch if desired
3 Connect power cord to outlet
4 Power on console
5 Allow console to fully initialize
6 Attach probe via connector to console
Main Screen Display
Ablate Power Settings
Coag Power Settings
9 Change settings using touch panel display
Device Recognition Indicator
Troubleshooting: Faults and Solutions
Unit does not power on
1. Check the main power plug and wall receptacle. 2. Check the main fuses per the User Guide instructions. 3. Call Arthrex® Tech Support at 1-888-420-9393.
Accessory ApolloRF™ probe does not work
1. Disconnect and reconnect probe. 2. Turn off console, wait 30 seconds, repower console. 3. If applicable, disconnect and reconnect footswitch, replace if footswitch not functioning. 4. Open new ApolloRF probe.
Yellow fault message
Refer to User Guide for proper disposition.
Red fault message (Critical Failure)
1. Make note of message displayed. 2. Call Arthrex Tech Support at 1-888-420-9393.
To return or repair device: Contact Arthrex Product Surveillance at 1-866-267-9138.
Arthrex, Inc. 1370 Creekside Blvd. Naples, FL 34108-1945 USA Toll Free: 1 (800) 934-4404 www.arthrex.com Arthrex Tech Support: +1-888-420-9393
Arthrex GmbH Erwin-Hielscher-Strasse 9 81249 München, Germany Tel: +49 89 909005-0 www.arthrex.de +49 89 90 90 05 8800
To Adjust Settings APOLLO 90 1. Press menu setting button 2. Press desired setting
For Language: Select desired language and then do one of the following: Press HOME to save setting and return to main screen -OR- Press Setting Button to save setting and return to MENU -OR- Press X to return to MENU without changing default
For Footswitch Override: Select desired option and then do one of the following: Press HOME to save setting and return to main screen -OR- Press Setting Button to save setting and return to MENU -OR- Press X to return to MENU without changing default
Button Activation 1. Ablation power setting 2. Ablate 3. Coag
For Volume: Select desired volume using + or – buttons and then do one of the following: Press HOME to save setting and return to main screen -OR- Press Setting Button to save setting and return to MENU -OR- Press X to return to MENU without changing default 1 2 3
©2016, Arthrex Inc. All rights reserved. LI1-00027-EN_B