This document offers helpful tips on using the Embletta® Gold™ system. For detailed information on using the device, consult
the Embletta Gold System Tutorial enclosed with your device, or the Clinical Manual located in the documentation folder of
the RemLogic-E™ Installation CD. To access the help file in RemLogic-E or RemLogic, press the F1 key. For more information,
please contact Embla Technical Support (firstname.lastname@example.org) or your Embla distributor.
When running RemLogic™ and RemLogic-E on multiple computers, ensure all are running the same program
version. To verify the program version, select Help | About from the application main menu.
For PAP studies without the ResMed AutoSet, remember to clear the AutoSet check box when preparing the
study in the Recording Wizard.
Sheet templates are customizable, reusable, pre-designed layouts for viewing recordings. To save
modifications to a template, right-click the sheet tab (located at the bottom of the sheet) > Select Update
Customizing Reports can be customized to hide/display report sections, modify axes and graphical display. To customize a
report, select Analysis | Settings from the application main menu > Select Custom from the Profile dropReports
down list > Expand the Reports Graphs and Report Sections nodes > Select an item to display the value
table > Double-click the appropriate cell in the Value column to modify the value.
The XFlow parameter is available for studies used with XactTrace™ respiratory belts. XFlow, a calculated
parameter, represents true flow. The Respiration Analyzer analyzes the XFlow signal similarly to the regular
flow signal from the nasal cannula. To use the XFlow in place of the nasal cannula signal, select
Analysis | Settings from the application main menu > From the Profile drop-down list, select Custom >
Expand the Apnea/Hypopnea node > Select Apnea or Hypopnea to display the value table > Double-click
the cell in the Value column corresponding to the parameter Select Trace Automatically > Select No.
The Embletta Gold comes equipped with a rechargeable battery and recharger. When fully charged, the
Embletta Gold will record for a minimum of 24 hours.
Medical tape can be used to fasten the cannula tube to the patient's cheeks. Attaching the Embletta Gold to
the patient before connecting the nasal cannula to the Luer Lock is recommended.
Sp0 2 Trace
Numerous SpO 2 trace artifacts might indicate a faulty oximeter probe. Replace the component to verify.
Failure to receive oximeter data might indicate a faulty connection. Verify the connection, and check the
Use the EKG profile in RemLogic or RemLogic-E.
Select EMG on the channel configuration screen. Record limb movement activity for the diagnosis of Periodic
Limb Movement (PLM) in addition to standard Embletta Gold parameters.
Select EEG on the channel configuration screen. Record 1 channel of EEG in addition to standard Embletta
For online studies, use only the Embletta 3-in-1 cable. RemLogic 1.1 or later is required.
As your total solution provider, we offer a complete selection of competitively priced sleep system sensors and
accessories at www.shopembla.com.
Embletta Gold Helpful Tips. Copyright © 2012 Embla Systems. All Rights Reserved. Document Number: D-0808-003, Revision 3.0.
Issued September 2012. Printed in the USA or Germany. Embla and Embletta are registered trademarks of Embla Systems. RemLogic, RemLogic-E,
and XactTrace are trademarks of Embla Systems. All other trademarks are the property of their respective owners.