Product Information: HR900 [LI900Bx]
BASIC HR900 INITIALISATION INSTRUCTION
This instruction is to be used in the event your HR900 bed functions do not work or you are experiencing flashing lights on
the Lateral Caregiver Unit [LCU]
If you are experiencing a flashing LCU or audible alarm, please proceed with Initialisation instructions, below.
Firstly, attempt to disable the attendant lockout controls on the LCU. If unsuccessful, initialise the system by 1 simultaneously
pressing the HEAD DOWN and THIGH DOWN keys, until the system stops beeping. Please note that the two buttons must be
activated at exactly the same time, held for approximately 5 seconds or until audible beep ceases.
If audible beep ceases, test functionality by 2 running HILO UP until both actuators are fully extended (bed is at its highest point). All
sensors and limit switches will now be Initialised (synchronised)