GENIUS™ 2 thermometer cleaning guide
Your GENIUS 2 thermometer should always be inspected prior to use.
Follow these simple step by step cleaning instructions to ensure maximum
performance from your GENIUS 2 infrared tympanic thermometer.
Always ensure your GENIUS 2 thermometer is clean,
especially the probe tip lens.
A dirty lens will reduce the temperature reading.
The lens must always be checked prior to use. It should
have a mirror-like finish, and be free from debris, fingerprints
If the lens remains soiled, clean with a cloth or cotton bud
dampened with warm water and mild detergent, or with
isopropyl alcohol wipes. Always allow 45 minutes drying
time with alcohol cleaning.
Return the thermometer to its base unit and allow upto
45 minutes drying time prior to using the thermometer if
alcohol has been used on the Probe Tip.
For routine lens cleaning, wipe with a lint-free cloth.
Always point the probe tip downwards when cleaning the lens.
Follow this cleaning by wiping the lens with a lint-free cloth to
remove any excess alcohol / detergent film.
Tyco Healthcare - Australia
166 Epping Road,
Lane Cove NSW 2066
Customer Service Ph: 1800 252 467
Customer Service Fax: (02) 9418 7137
To clean the handset and base, use warm water and a mild
detergent. Water temperature should not exceed 55˚C.
• Always use a new GENIUS 2 AccuSystem™ probe cover
for each temperature reading.
• Do NOT soak, submerge or rinse the GENIUS 2
thermometer under water.
• Never use an abrasive pad or an abrasive cleaner on the
GENIUS 2 thermometer.
Do not use alcohol on the LCD screen of the thermometer
as this may cause damage.
Tyco Healthcare - New Zealand
Ground Floor, 15B Vestey Drive,
Mount Wellington, Auckland
Customer Service Ph: 0508 489 264
Customer Service Fax: 0508 489 263
GENIUS™ and AccuSystem™ are trademarks of Tyco Healthcare Group LP or its affiliates.
© 2007 Tyco Healthcare Group LP. All rights reserved.
Therm 020-06-07 Genius 2 Cleaning
Infrared Tympanic Thermometry