Quick Reference Guide
NON-MYDRIATIC AUTO FUNDUS CAMERA AFC-330
Quick Reference Guide
Keep this manual handy for reference.
When the non-mydriatic auto fundus camera, AFC-330, is used, operate the device in accordance with this quick reference guide. If the device is not shut down with a proper procedure, device failure or corruption of data transferred to external devices such as a USB flash drive may result. For details, refer to the AFC-330 Operator’s Manual.
1. Before turning on power to the device, make sure that the power cord is securely connected to a grounded power outlet. ↓ 2. Turn on power to the device. The initial screen appears. Wait about 45 seconds.
↓ 3. The anterior eye observation screen is displayed.
1. Press the shutdown button
at the top left of the
anterior eye observation screen. ↓ 2. Press YES in the confirmation dialog box. In about 20 seconds, the message “Please turn off power.” appears.
↓ 3. Turn off power to the device.
Precautions Be sure to start up and shut down the device in the specific order. To turn on power to the device again, turn on the power switch at least 3 seconds after turning off the power switch. Be sure to connect or disconnect a USB flash drive with power to the device turned off or with the USB Flash Drive Settings screen displayed. In the USB Flash Drive Settings screen Connection: A USB flash drive can be connected when the USB Flash Drive Settings screen is displayed. Disconnection: In the USB Flash Drive Settings screen, select the USB flash drive to be removed, then press the eject button so that the USB flash drive disappears from the screen.
Be sure to check the USB flash drive for viruses before connecting it to the device.
Be sure to consult NIDEK or your authorized distributor when purchasing a USB flash drive to be used for the AFC-330. The driver of the USB flash drive needs to be installed in advance. Be sure to ask NIDEK or your authorized distributor for installation of the USB flash drive.
NIDEK CO., LTD. (Manufacturer)
: 34-14, Maehama, Hiroishi-cho, Gamagori, Aichi 443-0038, Japan Telephone: +81-533-67-6611 Facsimile: +81-533-67-6610
NIDEK CO., LTD. (Tokyo Office) NIDEK INCORPORATED (United States Agent)
: 3F Sumitomo Fudosan Hongo Bldg., 3-22-5, Hongo, Bunkyo-Ku, Tokyo 113-0033, Japan Telephone: +81-3-5844-2641 Facsimile: +81-3-5844-2642 : 47651 Westinghouse Drive, Fremont, California 94539, U. S. A. Telephone: +1-510-226-5700 Facsimile: +1-510-226-5750
NIDEK S.A. (EU Authorized Representative)
: Europarc 13, rue Auguste Perret, 94042 Créteil, France Telephone: +33-1-49 80 97 97 Facsimile: +33-1-49 80 32 08
Original instructions 15411-P972C February 2013
Image capture procedure The operating procedure below is explained in Normal photography mode using the auto shot without the blink detector and without auto switching to Small pupil photography mode that is the default setting. Patient list button
1. Remove the objective lens cap. 2. Press the patient list button in the anterior eye observation screen. The patient information register screen is displayed.
3. Press the desired patient for whom images are to be captured. Then
press the image capture button. To perform image capture for a new patient, press the patient register button at the bottom of the patient
Anterior eye observation screen
information register screen, and input the patient information in advance.
The anterior eye observation screen is displayed. The name of the selected patient is displayed at the top of the screen.
4. Prepare the patient for image capture. Image capture button Patient register button Patient information register screen
5. Perform alignment using the joystick so that the patient’s anterior eye is displayed on the LCD-touch screen.
6. Make the (eight) cornea-reflected spots appear.
When the auto tracking is activated, the fundus observation screen is displayed. When the auto focus is activated, the flash is automatically fired and an image is captured. The captured image is displayed in the preview screen.
Anterior eye observation screen 26
7. To save the image, press the OK/Auto transfer button.
The captured image is saved in the USB flash drive, and the review screen is displayed. 䎬䎧
8. To capture images of the other eye, pull the joystick toward the operator,
Fundus observation screen
䎬䎧 䎳䎷䎔䎔䎓䎙䎓䎛䎓䎓䎓䎕 䎤䏅䏆䏇䎃䎨䎃䎩䏊䏋
then move the image capturing unit sideways toward the other eye to display the anterior eye observation screen.
OK/Auto transfer button