Physical appearance check
Prior to the start of this process confirm that the unit has been cleaned
and disinfected correctly. A formal record of cleaning and disinfection
must be present and available prior to the start of the following activities.
Use gloves until cleaning/disinfecting is completed.
Cleaning and disinfecting
1 Open the shipping carton and place the pump in its plastic bag under the
2 Verify that the customer included his Declaration of cleaning and
disinfection as required. If cleaning cannot be verified do not continue,
3 Remove pump from plastic and discard bag.
4 Carefully clean all surfaces of the pump, power cord, canister holders
and universal holder (if returned) with germicidal/antibacterial treated
disposable wipe. Pay close attention to the handle and areas closest to
the location of the canister. For stubborn or larger debris use a soft bristle
5 Use a second wipe to thoroughly clean the surface.
6 Allow to remain wet for 5 minutes.
7 Dry surfaces with a disposable towel or cloth.
Do not immerse any part of the pump in fluid or use unnecessarily
Smith & Nephew, Inc.
970 Lake Carillon Drive
St. Petersburg, FL 33716
Customer Care Center
©2009 Smith & Nephew, Inc., All rights reserved.