L9000 Light Source
Run / Standby switch
Light Intensity Display
1. Turn on Power Switch
2. Make sure light cable release button is in unlock position
3. Insert light cable
4. Switch Stand-by control to Run when surgeon is ready to proceed with procedure.
5. Adjust brightness using Light intensity button.
6. Switch to standby at completion of case
Ensure lever is depressed
fully. Push light lead in firmly
until jaws engage.
At end of case ensure that
you depress the lever fully
when removing light cable.
For further information refer to the product IFU