To use CareFusion Surgical Clippers:
Wet clip whenever possible to help prevent airborne skin cells
and loose hair dispersal, which can contaminate the wound site.
Hold the skin taut with one hand while clipping with the other,
ensuring a smooth surface for clipping. This technique is especially
important when clipping loose skin areas. Avoid skin tags and
other inconsistencies with the skin’s surface.
Always rest the base of the clipping blade flat on the patient’s
skin. Do not change the angle or “toe in” the blade in an attempt
to get a closer clip.
Maintain a consistent grip on the clipper handle. Go over
the area thoroughly on the first attempt to ensure clipping
accuracy and precision.
Always move the clippers away from your body with a gentle
push. Do not pull the clippers toward you across the patient’s
skin, and never apply greater force than needed.
1. Attach the blade
With gloves on position a new blade
on the top of the clipper head. Align
the mark on the clipper blade to the
arrow. Slide the blade toward the dot to
secure it in place until you hear a click.
2. Clip the hair
Hold the clipper using a pencil
grip or natural overhand grip
with the CareFusion logo facing
up. Stretch the skin, rest the clipping
blade flat on the skin and clip against
the grain of the hair.
3. Remove the blade
Turn the clipper off, position the
clipper blade facing downward over
a sharps trash receptacle and use your
thumb to push the blade forward
along the blade frame.
4. Clean the clipper handle
Wash the clipper handle thoroughly
with soap and water, and then wipe
the device with surface disinfectant
or soak the handle in an antimicrobial
agent for no more than 30 minutes.*
Rechargeable surgical clipper
General purpose blade
To order, contact your local representative or call 800.523.0502.
CareFusion Surgical Clippers can be cleaned with a mild disinfectant or a disinfectant wipe.
Disinfectants containing benzyl alcohol can cause chemical damage to the external surface of
the clipper handle or charger. This could void the warranty on the device. Submersion cannot
exceed one meter in depth for up to 30 minutes.
It is recommended that clinicians unplug the charger from a power source when the solid
orange light is no longer illuminated, indicating the device is fully charged.
© 2013 CareFusion Corporation or one of its subsidiaries. All rights reserved. SensiClip, CareFusion and the CareFusion
logo are trademarks or registered trademarks of CareFusion Corporation or one of its subsidiaries. SU2128 (0713/5000)
5. Charge and store
Place the clipper handle in an upright
position in the charging adapter. Lithiumion battery technology facilitates longer
runtime and improves energy efficiency,
and a battery life indicator light alerts
clinicians when to charge the unit.**
Instructions apply to all CareFusion Surgical Clipper blades.